Sequencer Command Documentation
Go to the SEQUENCER Reference Manual Main Page.
||[-d] [ready] [step] <n>
||Moves ALFOSC internal focus to step position <n>. [ready] Returns '0' if the element is moving, '1' if it is ready. [step] returns the step position [0-2700] of the ALFOSC internal focus.
||The command instructs the ALFOSC instrument user interface (UIF) to re-read the setup information. The corresponds to pressing the "read setup" button in the UIF.
In scripts, the syntax alfosc.<command> should be used.
If a part of a command is written in <angle brackets>, this is an argument that should be replaced with a value.
If a part of a command is written in [square brackets], this is an optional argument or flag.
When available, the optional flag -d will write extended debugging information to the log.
When available, the optional flag -e will show the script on standard output instead of executing the commands.
When available, the optional flag -t will only make a simulated exposure. Useful for testing scripts.
DATA: All commands related to handling/modifying data (editing fits-headers, post-processing etc.).
DET: Commands related to operations with the detector.
DISP: Commands related to displaying images on the instrument computer.
INST: Commands related to operations with the instrument.
TEL: Commands related to operations with the telescope.