M1 and M2 aluminisation May 2002
Both mirrors were realuminised on the 27th of May 2002. The previous
aluminisation took place in May 1997 for M1 and June 1995 for M2. The
started by dismounting the mirrors on the 24th of May, and the mirrors
were transported back to the NOT on the 29th of May. While the mirrors
were out, the telescope center section was machined to improve roundness.
Alignment was done on the 30th, and a report on the
was produced by Anton Sørensen, CUO.
The throughput was tested with ALFOSC and StanCam on the 31st of May.
results for ALFOSC show the following sensitivity improvements:
U_after / U_before = 1.59
B = 1.51
V = 1.36
R = 1.32
I = 1.29